The oxygen index tester is used to determine the concentration of oxygen in the polymer during combustion (volume percentage).his tester can be used as a means of identifying the combustion performance of polymers, but also as a related research tool, so as to obtain a better understanding of the combustion process of the polymer. It accords to ASTM D2863、ISO4589-2
The oxygen index tester is used to determine the concentration of oxygen in the polymer during combustion (volume percentage).
The polymer oxygen index value is the minimum oxygen percent by volume in oxygen & nitrogen mixture gas, which can keep the material burning 50mm or keep burning 3 minutes.
This tester can be used as a means of identifying the combustion performance of polymers, but also as a related research tool, so as to obtain a better understanding of the combustion process of the polymer.
The instrument is suitable for plastic, rubber, foam, wood and other materials, the combustion performance of the test, the results are accurate, good reproducibility. It is an indispensable test equipment for the research and production of flame retardant materials.
This tester uses the German ENVITEC high precision oxygen sensor, digital display results, high precision, long service life, simple structure, convenient operation, no need to calculate by formula. Front panel is ash black plastic surface scrub treatment, 304 stainless steel decorative strip surrounding, the whole machine spray plastic, appearance beautiful, generous.
Main technical parameters:
Burning barrel inner diameter: 100mm
Burning barrel height: 450mm
Flow meter accuracy: 2.5 class
Pressure meter accuracy: 2.5 class
Pressure meter resolution: 0.01Mpa
Oxygen indicator resolution: 0.1%
Gas source: Oxygen, nitrogen purify not less than 98%
Test condition: Temperature: 10-35℃ Humidity: 45-75%
Input pressure: 0.2-0.3MPa
Working pressure: 0.05-0.15MPa
Sample type: Self-supporting material, Non self-supporting material
Power: 110V, 60HZ, single phase
Packing size: 630*730*800mm
Gross weight: 58Kg
1. The internal pressure of oxygen and nitrogen cylinder used by the tester is not less than 1MPa. If it is lower than this pressure, please replace it in time.(Need user prepare Pressure relief valve,Oxygen and nitrogen themself)
2. adjust the flow when the force should be uniform, do not suddenly big or small.
3. according to the use of regular cleaning of the combustion cylinder and ignition surface dirt, timely removal of burning cylinder metal wire burning drops, to ensure gas flow.
4. in case of problems, please contact our company, please do not disassemble, especially components.
5. oxygen, nitrogen cylinder belongs to high pressure container, use must be far away from the fire source, pay attention to safety.